Necklite – Moldable Neck Brace for Adults Yellow
- 125.00 Grams
- Calculated at Checkout
- Compact, light, fast and easy to use.
- Cervical stabilization for extrication and rehabilitation
- Large tracheal opening allows for optimal visualization
- Helps reduce lateral movement
- Foldable and moldable before and after application
- X-ray translucent.
- The locking system uses a 5 cm black Velcro strap.
- Just place and close on patient.
- Non-rigid, latex free.
- Dimensions: Width 550 mm – Height 180 mm – Thickness 5 mm
Introducing a moldable emergency neck brace, comprised of a soft aluminum and coated with high density PU foam. These specific design characteristics allow the brace to be folded or rolled for compact storage. This allows for convenient transport in backpacks, kits and narrow pockets. A series of moldable small wings are positioned under the chin in order to conform to each patient’s unique features. Once applied to a patient, it becomes rigid and the neck is tightly immobilized. Once in place, the device features an elliptical hole on the backside and a large oval shaped hole on the front side which allows for a tracheotomy. Necklite is X-ray translucent, it allows therefore to submit the patient to X-ray while he wears on the device. CE approved.
Necklite- Adult - Yellow/Grey
A one size fits all moldable cervical stabilization device.
Attitudes toward the use of different types of cervical collars. Experimental study on healthy volunteers
Cervical spine immobilization in case of suspicion of spine injury is an important element of rescue procedures aimed at protecting the patient against a potential worsening of injuries. The current trend is away from the routine use of the cervical collars in cervical spine injury. Researchers indicate, among other things,the occurrence of pain resulting from the use of cervical collar, as well as difficulties in performing endotracheal intubation when a standard collar is used. The aim of the study was to assess attitudes towards the use of different cervical collars in an experimental study involving healthy volunteers.
The study was attended by 53 medical students who were put on 3 types of cervical collars by experienced paramedics. The cervical collar was worn for 10 minutes, followed by a 30-minute break and a different type of collar. The following types of cervical collars were used in the study:
1) Stifneck® Extrication Cervical Collar (Stifneck; Laerdal, Stavanger, Norway),
2) Philadelphia Patriot One-Piece Cervical (Patriot; Össur, Reykjavík, Island),
3) Necklite – Moldable Neck Brace (Necklite; FLAMOR SRL, San Pietro Mosezzo, Italy).
Both the order of participants and methods of cervical spine stabilization were random. The coin flip technique was used for this purpose. The analysis included pain related to the cervical collar insertion, assessed on a scale from 1 to 10, where "1" meant no pain, and "10" - increased pain. Another parameter was the pressure on mastoid process, also assessed on a 10-degree scale, "1" - no pressure, and "10" - increased pressure.
The study involved 53 paramedics whose median age was 34 (IQR; 29-40) years. The severity of pain in Stifneck, Patriot and Necklite collars was differentiated and amounted to 4 (3-6) vs. 5 (3-7) vs. 1 (0-2) points, respectively. The level of pressure on mastoid processes in the Stifneck collar was 5 (4-6) points, 5 (5-7) points for Patriot collar, and 1 (0-1) points for Necklite. Significant statistical differences in pressure on mastoid processes between Stifneck and Necklite (p<0.001) as well as between Patriot and Necklite (p<0.001) were observed.
The study showed that the use of Necklite Moldable Neck Brace neck collar, due to its innovative design, was associated with the slightest pressure on mastoid processes and the reduction of pain associated with cervical spine immobilization.
Conflict of interest
Nothing to declare.